Search the Wiki
Hydrogen Fine Structure
The great success of the Dirac equation, which naturally predicted the spin and magnetic moment of the electron, spin-orbit interactions, and features of the fine structure of atomic spectra, had nevertheless an important incorrect prediction: that electron states with the same total angular momentum j should have the same energy. For many years, good measurements of the hydrogen fine structure that could test Dirac's theory remained elusive because Doppler broadening—the spreading out of atomic spectral lines due to the motion of the atoms—made it impossible to see the fine structure clearly until the work by Lamb & Retherford. They circumvented Doppler broadening by using microwaves to induce transitions between fine-structure states directly and they proved convincingly that the predicted Dirac degeneracy was not there, but instead measured a shift between the two lowest excited states, 2S1/2 and 2P1/2, of about 1000 MHz. This became known as the “Lamb shift,” and it was a key experimental finding that motivated the development of quantum electrodynamics of Feynman and Schwinger.
With the development of tunable lasers, a new technique of absorption spectroscopy that allows the hydrogen fine structure to be observed directly became possible. This technique uses crossed beams, with each beam traveling in opposite directions through a gas discharge tube filled with glowing hydrogen. Only those atoms that have zero velocity along the beam axis can interact with both beams, and this causes the signal from one of the beams to show a small Doppler-free feature on top of the Doppler-broadened spectrum. By chopping the other beam, and thus turning such features on and off, one can selectively amplify these small features, and obtain a Doppler-free spectrum.
In this Immersion, participants will learn this technique of crossed-beam Doppler-free absorption spectroscopy, and use it to measure the fine structure of the Balmer-α line (at 656.28 nm), and in so doing measure the Lamb shift. Thus, this experiment gives one the opportunity to learn an experimental technique that is in current use in many research laboratories, and also to measure one of the most important experimental results that support modern quantum electrodynamics, arguably the king of 20th century physical theories for its success in explaining the hydrogen spectrum to extraordinary precision.
This experiment uses the most complex and specialized setup we have in the undergraduate laboratories. At its center is the gas discharge absorption cell and housing box. The absorption cell is custom built by our glassblower and allows laser beams to pass along about a 50 cm length occupied by a glowing hydrogen discharge. The tube and plumbing is designed to allow a continuous flow of hydrogen, and the gas is lit by a 2000 V DC supply. The laser is a New Focus model 6202 external-cavity diode laser acquired in 1999, specifically designed to scan through the Balmer-α line. A system of mirrors, lenses and a beam splitter are used to separate the laser beam into a bright “pump” beam and less intense “probe” beam, with each beam passing through the discharge tube in opposite directions. Finally the pump beam is chopped, and the synchronous signal is used to eliminate noise and background from the signals picked up by the probe beam detector and another detector at the output of a Fabry-Perot cavity. A dual-pen chart recorder captures the sweep signal from both detectors. By comparing the probe signal with the Fabry-Perot signal, an accurate measurement of the separation of spectral lines can be carried out. Signal generators and interconnecting boxes are used to rout signals and drive the laser sweep electronics. In the pictures above are shown the discharge tube and an overview of the optics layout, plus the equipment rack containing a lock-in amp, HV supply and signal generators. The inset at lower right shows the lamp in its "on" state.
Laser absorption spectroscopy is a current technique in active research. This experiment illustrates the technique extremely well and includes a vivid example of the power of lock-in amplification to separate signal from background.
Among the experimental topics to be covered are understanding and using tunable external-cavity lasers, operating hydrogen gas-discharge absorption cells, layout and alignment of crossed-beam optics, measurements of frequency shift with Fabry-Perot cavities, and using optical choppers and lock-in amplification to extract tiny signals from a large background. Sufficient theory will also be studied to provide the participant with ways to teach the concepts to undergraduate physics students.
Outline of the Immersion:
All essential materials and equipment will be provided, but participants are encouraged to bring a camera (cellphone OK) and computer/notebook to record notes and results. If the participant has a favorite data analysis package, that will be useful during the lab.
There are a few hazards in this experiment: a high-voltage power supply gas discharge lamp; UV radiation from discharge tube, hydrogen gas, and a medium-power (5-10 mW) laser (Class IIIb). Instruction will be provided concerning these risks.
As noted, this experiment is one of the most advanced in our department, and the components are not cheap or simple to obtain. Our setup was mostly completed in 2000, so the following prices are at best a rough guide:
brought to you by the AAPT, APS and NSF-NSDL
a member of the comPADRE Digital Library
©2007-2017, All Rights Reserved
contact Advanced Labs
|about - disclaimer - terms - privacy - faq - sitemap|